GEA- MK2.1- L-PROBE WITH THERMOELEMENT
- Oxygen measurement in gas atmospheres
- simple gas sampling
- Insensitive to temperature
- Insensitive to vibration
- Easily replaceable
- Fitting adapted to customers needs
The probe measures the residual oxygen content in a measuring chamber at the end of the measuring gas fitting.
The probe signal can be tapped off as a voltage. The probe is heated with a constant voltage. For supplying the heating only power supply inits approved by MESA may be used.
The quantity of gas passing the probe should be within a range of 20…50 l/h. Too much gas volume can change the highly constant probe temperature, which results in a falsification of the measured value. Every probe includes two constants (K1 and K2). These constants must be entered in the measuring instruments (e.g. intelligent measuring transmitter Carbo47,Carbomat-M, MCON Carbo).Ifnottherewillbe some measuring faults.
|Dimension of sensing head:||105 x 290 x 45mm (BxHxD)|
|Rated length:||400…1100 mm|
|Connection L-probe:||snap closure|
|Necessary additional parts:||NTV44P for heating the L-probe and controlling temperature.|
|Measuring range:||0…1300 mV|
|Optional devices||• Carbo 15/ 47
• MCon Carbo 16-48
• Carbo 100
• Carbo 1000
• MCON Carbo
83075 Bad Feilnbach